Instrument is without sample, without reference. After cleaning procedure are 3-4 followings base line correct, follow-up base line has the peak in this temperature range.
This action is repeating.
If is measuring of standards, In, Zn on the "correct" base lines, results are OK.
Thanks for the technical advice.
Reply to Jaroslav Smejkal RE: PerkinElmer - DSC 8500