The Varian CP-3800 GC meets all your analytical requirements with practical solutions for simple and complex applications. In a standard configuration, the CP-3800 accommodates up to three injectors and three detectors, all operating simultaneously. In addition, the CP-3800 is easily configured to perform on-line analyses for monitoring critical gas and liquid process streams. The CP-3800’s flexible platform lets you start with a basic configuration and add a wide variety of options in the field. Options include additional channels, gas and liquid sampling valves, and external sampling devices. Future upgrades guarantee that you can meet changing requirements at a moment’s notice. Unleash the full potential of the CP-3800 GC with the Galaxie Chromatography Data System. Galaxie dramatically improves daily workflow with intuitive, easy-to-use graphical interfaces for instrument control, data acquisition, processing, and standard and custom reporting. From a simple stand-alone GC system, to an automated Simulated Distillation package, the CP-3800 GC with Galaxie provides a powerful solution to all of your GC analytical needs. The CP-3800 GC from Varian—unlimited possibilities!
|Temperature Range||4 to 450 °C|
|Detector||Flame Ionization Detector (FID) • Thermal Conductivity Detector (TCD) • Electron Capture Detector (ECD) • Thermionic Specific Detector (TSD or NPD) • Pulsed Flame Photometric Detector (PFPD)|
|GC System Type||Modular|
|Humidity||5% to 95%|
|Number of Detectors||3|
|Power Requirements||101V, 120V, 230V|
18 days ago
|Replase part numer for TCD |
Please guide to the replacement of the TCD detector GC varian CP 3800 with a similar agilant that is also resistant to H2S. Thanks
a month ago
|Varian CP-3800 syringe volume exceeded" problem|
The autosampler on my CP3800 GC system suddenly developed a "syringe volume exceeded" problem:
"8400 Autosampler (Front) 84x0 syringe volume exceeded"
Does anyone know how to fix this?
2 months ago
|equilibration status |
i have a varian 3800 GC, my stabilization time is set at 2 mins but my equilibration time is on for the past 4 days. what can i do to correct that