The Titration Excellence line offers many advantages such as One Click Titration™, Plug & Play and method database. The T50 is the ideal entry-level model for the Titration Excellence line and offers much more than a simple titrator:
Terminal with brilliant Touch Screen Expandable with dispensing burette drives Comprehensive communication connections Preconfigured system application packages
3 months ago
|Poor Accuracy |
We are using 0.1N Silver Nitrate to analyze Chloride. We run a daily system suit check with 0.1N NaCl as our "standard" to measure the system. Starting last week we began having issues with the readback running high at ~0.1015N but it was consistently doing this - %RSD with 3 replicates less than 0.5% RSD. We are now seeing the issue swing the other direction - readback is now closer to 0.098N Silver Nitrate but again it is consistently reading three replicates with low %RSD. There is a small amount of silver nitrate at the top of the burette indicating a small/slow leak because it was noted a while ago by another analyst while we were waiting for a PO to go through for a service visit. Is it finally to the point of not functioning due to a leak? But if so, why would it yield such consistent results? I would expect a leak to give results with more variance. (Also to note we tried a new silver electrode yesterday with the same results - granted it had been sitting in a box so that brings up another question of how long does a new electrode (or one that has been sitting) need to sit in the electrode solution before use and is there a lifetime in regards to the liquid inside the electrode and how often should it be replaced?)
3 months ago
|MF  Undefined position|
Does anyone know what MF  Undefined error means on the T50. I got this error once I started to run my samples.
4 months ago
| Storage Medium Not Found|
I have this Error Message displayed :Storage Medium Not Found
Can anybody assist on this.