3 months ago
I would like to know more information about San++ on how to analyze cyanide, surfactants and phenols. How to calculate reagent expenses, analysis time and technical procedures.
a year ago
|PROBLEMS WITH PEAKING|
Hello there my skalar san++ analyzer recently doesnt detect the peaks of samples analyzed. the reagent lines have been changed and also cleaned as per the maintenance proceduer but the peaks not only are dimished but do not show at all. how should i go about this?
a year ago
we are currently using skalar at a remote location, and about two days ago, we are having trouble with the baseline. TCN always stay about 1.2-1.3 and WAD 1.6-1.7. But now TCN is sitting at 2and WAD is at 3. The baseline is also unstable, with blip like picks all over, thats before using any in house prep solutions. only using DI water. what will be the possible issue here?, we are pretty sure that all solutions lines are flowing well and the glass coils are also clean. We have limited information on possible issues to encounter with the instrument. Can you please help us trouble shoot any possible cause that can affect the baseline and the instrument performance.