7694 Headspace Sampler
# It can eliminate sample preparation by letting you introduce volatile components automatically from virtually any sample matrix directly into a GC # Integrates seamlessly with HP/Agilent GCs # Features a highly inert path from the vial to the GC column head to prevent sample absorption, improve peak shape, maintain sample integrity, and enhance sensitivity # Automated 44-sample tray # Constant heating time # Overlapping sample processing; automatic parameter increments for easy method development # Up to 10 headspace extractions per vial # Two levels of vial agitation
a month ago
|7694 HS with 6890 GC/5973N MS|
I am trying to connect an Agilent 7694 Headspace sampler to a 6890GC/5973N MS system. The gas line connection seems pretty straightforward but I am not sure about what types of cables/connections need to go from the Headspace to the GC. I have been trying to decipher the manuals but they are not as helpful as I had hoped. Can anyone help ?
2 years ago
|CHANGING DEIONIZED WATER |
NEED A PROCEDURE TO CHANGE OUT DEIONIZED WATER
2 years ago
|Autosampler Down--Manual insertion?|
Our autosampler is currently experiencing problems with one of the potentiometers in the main carrier belt. While it is out for repairs, is there a way to manually insert the vial into the port and tell the GC to drop the vial and begin running it? I've heard rumors of a mysterious button you can push, but I can't seem to find it.