The Thermo Scientific* TriPlus RSH Autosampler Enables Flexible, Automated Sampling for Ultimate Productivity in GC and GC-MS Analysis The Thermo Scientific TriPlus RSH autosampler expands automated capabilities beyond sample injections to advanced sample handling cycles. Automating the whole workflow, from sample pipetting to injection, increases precision and reproducibility and enables unprecedented flexibility and productivity in GC and GC-MS sampling.
Innovative Automatic Tool Change (ATC) Enables seamless operation by automating the exchange of syringes for different tasks in a single, unattended sequence prior to sample injection High precision sample-handling capabilities eliminate the need for extensive manual labor and downtime traditionally required to complete tasks, such as standard or stock sample dilutions, internal standard addition and derivatization Automatic switching of syringes for different analytical tasks guarantees zero carryover Unprecedented Sample Capacity Expands unattended operations and productivity Unequivocal lab throughput is attainable by using the largest sample capacity available for an autosampler Sample vial capacity of 648 2-mL vials, combined with multiple 100-mL wash/waste bottles, allows the TriPlus RSH autosampler to achieve week-end long unattended operations—a goal not attainable with most other sampling systems currently on the market Vial Bottom Sensing Expands application range of TriPlus RSH autosampler Precisely detecting vial bottom enables the autosampler to accurately withdrawn microliters, or fraction of microliters, from a sample as low as 5µL in a vial Examples of applications benefitting from this breakthrough include: trace analysis, radioactive samples, and samples requiring expensive internal standards Sample injection modes: Liquid injection (new Fast mode, injects in less than 100 ms; regular mode and Sandwich mode) Headspace (automated standard addition capabilities, optional MHE) Automated SPME Liquid Handling operations: Sequential dilution Calibration dilution Standard addition Derivatization Mixing Vortexing
7 years ago
Hello , my triplus autosampler is not responding to my instrumental method. For instance, the triplus autosampler immediately injects the next sample right after injecting the first one which is not my instrumental method, Please what should i do.